The world's ONLY ParametricRelease™ Instrument
If you are an InstrumentManufacturer (OEM)
click here for more information
1. Cleans - replaces manual cleaning
3. Rinses off germicide residue without compromising disinfection integrity
2. Disinfects to >10-6 no CFUs
Langford IC Systems, Inc
It is also designed to operate as the world'sonly Instrument Processor for lumened and
most other medical/surgical instruments in
a single chamber.
It obsoletes current washer devices, manual cleaningand liquid chemical AER devices
3/13/2014: We have received 2 new patents :#8,568,666,
These patents extend our protection
Draft FDA Guideline for Quantifiable Validated Cleaning
LIC can assist Instrument Manufacturers in qualifying their instruments as reusable* LIC can provide FDA approved protocols and test methods for validated cleaning.
* LIC will label it's device for cleaning of their instruments.
* LIC will provide the OEM with cleaning reprocessing instructions for their labeling.
* OEM can test the cleaning of the instrument early in the development process and make
the necessary design changes before production and FDA regulatory testing.
* LIC's device provides the manufacturer with the means to clean and validate in the final
manufacturing step before packaging and sterilizing.
Dramatic change in FDA policy regarding the Requirement to qualify as Reusable: developworst case soiling and cleaning endpoints using AAMI TIR 30 as the FDA suggested
Reusable Instrument Manufacturers (OEMS)
Cleaner only Device
It is the perfect support for
preparing lumened and
non-lumened instruments for
Sterilizing by cleaning to the
quantifiable validated level
reuired by the FDA.
Cleaner Ozone Sterilizer
Cleaning and sterilizing lumened
and non-lumened instruments.
The LIC Parametric Release ™ Instrument Processor System is unique in that the technology eliminates the need for connectors and is designed to process a variety of medical instruments including but not limited to endoscopes in a single chamber. It is the first and only Parametric Release liquid chemical Instrument Processor System that delivers the three critical steps required to process a reusable medical instrument, i.e. integrates cleaning to less than 2 micrograms of protein per square centimeter, sanitizes to greater than 10-6 with no surviving microorganisms and rinses the germicide residue with sanitized rinse water without compromising the sanitizing step.
Rinse off germicide residuewithout compromising the
4) The process includes a patented waterpurification system that provides a final rinse
of sanitized water.
Disinfect to >10-6 Nosurviving CFUs
The uniqueness of this
technology is not only
its ability to clean the
inside of lumens but
that it generates a
Shear that removes
the toughest soils
from the inside and
outside of an
All thisWITHOUT THE USE OF
5) Because we do not use connectors, the machine is designed to do general
surgical instruments and implantables, etc.
in the same chamber.
We have the only liquid chemicalcleaner processor that can:
1) The instrument processor does not useconnectors to attach to the instrument being
processed in order to direct the flow of
chemical detergent and germicide, so that all
external and internal surfaces are exposed to
the detergent and germicide.
2) No brushes are used that may scratch andscore the inner or outer surfaces of the device
Video of surge actionin the Instrument Processor
Clean to a standard of <2µg cm2of protein. (endoscopes)
3) The system is equally effective at cleaning &disinfecting all sized lumens, ranging from a
catheter to a colonoscope.
Mock lumen before being processed in theLangford Processor. The lumen is 4.5mm diameter
and 182 cm long. Note that the Hucker's soil
goes the entire length of the 4.5mm diameter lumen.
Same Mock lumen after being processed inthe Langford Processor. Note that the entire
length of the 182cm x 4.5mm lumen is clean.
Hucker's soil: a simulatedfecal matter acknowledged by
the industry as the toughest
182cm x 1.5 mm long Mock lumen with a 1 mm diameter end restrictor soiled with Hucker's
soil before being processed in the Langford Processor.
Before and After pictures showing internal cleaning of mock lumens:
Same Mock lumen with restrictor valve after beingprocessed in the Langford Processor.
Note that the entire 182cm x 1.5mm lumen is clean
including the 1mm end restrictor.
The chamber, baffle and tray are heavily contaminated with Hucker's soil. Not only were these mock lumens soiled internally and externally; all the surfaces of the chamber, the baffle and the tray top and bottom were also soiled.
Our processor safely and effectively cleaned the Hucker's soil from all the contaminated surfaces.
Washers merely agitate the fluid whereas our processor producesmechanical shear that strips the toughest soiling from surfaces.
This mechanical shear is developed at 12.5 psi.
OBJECTIVE: Evaluation of a proposed cleaning process for a fully intact Trocar utilizing the LIC Instrument Processor to determine if a Trocar can be cleaned without disassembling it.
BACKGROUND: A recent design change of the J&J Trocar, the Endopath Xcel, incorporatesan internal foam pad that has been a
challenge to clean both because of it's
surface texture and its isolated location
within the Trocar.
A further challenge that has been imposed is the excessive amountof organic soil (composed of horse serum, defibrinated sheep's
blood, and subjected to salt pork) which has been placed inside the
Trocar. The Trocar was subsequently baked in an oven at 60 0
Centigrade for 72 hours. This excessive amount of dried organic
soil interior to the Trocar has been observed and photographed by
disassembling another Trocar inoculated in the same group. After baking,
the Trocar sat for approximately 10 weeks prior to this feasibilty cleaning
Feasibility Evaluation report SummaryTrocar cleaning process
The testing demonstrated that the soiled fully intactEndopath Xcel trocar is an acceptable candidate
for cleaning using the LIC Instrument Processor
System. The Endopath Xcel Trocar was cleaned
to a level where no visible trace of any soiling
was detectable after disassembly of the separate parts of the trocar. Succesfuly proving that using the LIC Instrument Processor negates the need to disassemble and reassemble the Endopath Xcel Trocar in order to clean it for sterilization and eventual reuse. Thus proving that disassemblingand cleaning of each individual part of thetrocar is not necessary when it is cleaned in the LIC Processor.
The LIC Instrument Processor System cleaning
cycle is capable of removing all visible evidence
of the test soil and therefore based on prior
experience with similar devices should exhibit
≦0.60 µg/cm2 of residual protein when subjected toa protein assay.
Cleans to FDA standard(<2 µg/cm2 protein)
without requiring manual
Our Cleaner Processor is the ONLY System able to reliablyand verifiably sterilize patient-ready instruments.
Sterilizes to >10-6 , nomicroorganisms surviving
Verified sanitized waterrinse system
To illustrate the superiority of our
device over others in
the marketplace, wehave put together thischart to show ourdevice compared tocurrently used devices.
Designed to reprocesslumened and non
83 plus 2
Langford IC SystemsParametric Release ™
Steris System 1e
J&J ASPSterrad NX
Above the bar products
J&J ASPSterrad 100
Below the bar products
83 plus 9
Bob McCoyVice President
Any inquiries should be directed to:
Ron MaddixVice Presdient Marketing
310 S. Williams blvd.
Tucson, Arizona 85711